What is the recommended equipment for HDMI sink testing per CTS v1.3a?
Recommended Test Equipment for HDMI Compliance Testing per CTS (Compliance Test Specification) V1.3c
HDMI Sink Test
|Oscilloscope||DSA70804, DSA71254, DSA71604 or DSA72004 with Opt 2XL; or|
DPO70804, DPO71254, DPO71604, or DPO72004 with Opt 2XL
|Other Supported Oscilloscopes 1,2||TDS6804B, TDS6124C and TDS6154C||1|
|Software||TDSHT3 - HDMI compliance software (support for CTS V1.3b requires software Version 3.3.0 or higher. Support for CTS V1.3b1 and support of 4-channel measurement requires software Version 4.0 or higher)||1|
|Probes||P7313SMA differential SMA probe - use for all differential and single-ended measurements||2|
|Data Generator||DTG5334 with (3) DTGM30 modules||1 set|
|Other Supported Data Generators3||DTG5274 with (3) DTGM30 modules can be used but will only support HDMI testing at limited resolution and clock/data rate||1 set|
|Jitter Generator||AWG7122B with opt 01 - supports both standard (combined) and Optional (separate clock/data) jitter insertion methods||1|
|Other Supported Jitter Generators||AFG3102 - this support only the standard (combined clock/data) jitter insertion method||1|
|TDR System||DSA8200 with (1) 80E03 and (1) 80E04 module (for SINK Differential Impedance test)||1|
|Test Fixtures (TPAs)||ET-HDMI-TPA-S - HDMI Type A fixture set for Transmitter Test, Receiver Test, and Cable Test. Order from Efficere Technologies.||1 set|
|Connector||Bias-Tees to shift AWG7122B direct clock output to TMDS levels. Order individually as ZX85-12G-S+ from Mini-Circuits||2|
|Adapter||BNC to SMA adapter||1|
|Interface||NI-GPIB-USB -A/B/HS with Driver Software from National Instruments. Type B/HS is recommended for use with DPO/DSA70000 Series oscilloscopes.||1|
|Power Supply||Power supply 5V||1|
Transition Time Converters from Picosecond Pulse Labs
Please note that 5915-100-xxx type filter has Female connectors (jack) on both input and output sides. One Male-Male (plug-plug) adapter will be needed for each TTC unit to be connected to DTG output or cable emulator unit directly. Unit with different shape of connectors Female-Male (jack-plug) is available by ordering as 5915-110-xxx, but will be less flexible in connection.
1 Oscilloscopes with bandwidth less than 8 GHz will only support HDMI testing at limited resolution and clock/data rates.
2 16M per channel or greater record length is required for HDMI Compliance test.
3 If the Jitter Generator used is an AFG3102 or an AWG710B, then a single DTGM32 module is also required for use with either DTG model.
Эти часто задаваемые вопросы относятся к
Серия приборов не найдена
Идентификатор часто задаваемых вопросов 53541Просмотреть все ЧЗВ »