What is the difference between the P7513A and P7513?
The difference between these two probes is a longer cable on the P7513A. The added length offered the ability to probe circuits at a greater distance from the oscilloscope.
The P7513 had a cable length of 1 meter and a propagation delay of 4.4 ns +/- 0.1 ns.
The P7513A has a cable length of 1.3 meters and a propagation delay of 5.76 ns +/- 0.1 ns
This FAQ Applies to:
Product Series: Low Voltage Differential Probes
Product: P7513, P7513A
FAQ ID : 773521View all FAQs »