- The IConnect software assumes that the very first point in the measurement window is 50 Ohm, which is the default value for the Impedance Box in the Z-Line Computation Window. If you measure that first impedance point on the oscilloscope, and get a number slightly different from 50 Ohm, you should enter that number in the Z-line computation window. If you measure, for example, 50.4 Ohm in the oscilloscope, that can easily account for the difference between 53 and 55 Ohm towards the end of the waveform.
- Use an average impedance reading, never a point reading. Point reading is too susceptible to the local noise on the waveform. You can measure the average value in IConnect by utilizing our Single-Line modeling window. Load the impedance profile into that window, select the segment over which you want to read the impedance, and select Mean value in the Impedance drop down box on the right.
This FAQ Applies to:
Product: CSA8000B, CSA8000, TDS8000, TDS8000B, CSA8200, TDS8200, 80SICON, 80SSPAR, 80SICMX
FAQ ID : 56676View all FAQs »